The instrument that measures the low value of power factor accurately is known as the Low Power Factor Wattmeter (LPFW). The low power factor meter is used. So LPF(Low Power Factor) Wattmeter is used in OPEN CIRCUIT TEST of Transformer. In short circuit test the secondary winding of transformer. LPF wattmeter is used for measuring power in highly inductive circuits & UPF wattmeter is used for measure power in resistance circuit in which the power factor.
|Published (Last):||23 March 2008|
|PDF File Size:||5.60 Mb|
|ePub File Size:||7.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
Electrodynamometer wattmeter and Low Power Factor LPF wattmeters are mostly similar in construction and operation, except for the following modifications are carried out in the electrodynamometer wattmeter to convert it into a LPF wattmeter: As the name suggests the low power factor meter are the instruments that measures lower values of power factor accurately. There is a requirement of low power factor meter because, the usage of ordinary electrodynamometer wattmeter to measure power factor of a low pf load gives inaccurate results.
Now there are two main reasons that would suggests us that we should not use ordinary wattmeter in measuring the low value of power factor.
The pressure coil circuit is wattmeter to have a low value of resistance, so that the current flowing through it is increased to give an increased torque. Compensation for Pressure coil current: In this category two cases arise and their diagrams are shown below: In the first category both the ends of the pressure coil is connected to supply side i.
The supply voltage is equal to the voltage across the pressure coil. Thus in this case we have power shown by the first wattmeter is equal to the power loss in the load plus power loss in the current coil. In the second category, the current coil is not in series with the load and the voltage across the pressure coil is not equal to the applied voltage.
Answers and Solutions
The voltage across pressure coil is equal to the voltage across the load. In this power shown by dattmeter second watt meter is equal to the power loss in the load plus the power loss in the pressure coil.
Therefore, it is absolutely necessary to compensate for the pressure coil current in a LPF wattmeter. The compensating coil is connected in series with the pressure coil circuit and is made nearly identical to the current coil. The compensating coil is so connected that it opposes the field of the current coil.
Thus, the resultant field is due to current I only. Hence, the error due to pressure coil current flowing in current coil is neutralized.
Low Power Factor Wattmeter
Compensation for Inductance of PressureCoil: The error caused by pressure coil inductance is. The compensation of the error caused by the inductance in the pressure coil is done by connecting a capacitor across a part of series resistance in the pressure coil circuit.
LPF wattmeters are designed to have a small control torque so that they give full scale deflections. Incorporating all these modifications into an electrodynamometer wattmeter gives the following circuit: A modern low power factor meter is designed lppf that it gives high accuracy while measuring power factors even lower than 0.
Your email address will not be published.
Notify me when new comments are added. This site uses Akismet to reduce spam.
What is Low Power Factor Wattmeter? – Definition & Explanation – Circuit Globe
Sattmeter how your comment data is processed. Subscribe to our mailing list and get interesting stuff and updates to your email inbox. Why a LPF Wattmeter is needed? Leave a Reply Cancel reply Your email address will not be published.